Replacement water filters, RO reverse osmosis membranes and whole house filter systems. Refrigerator water filters, pool and spa filter. Pentek, OMNIFILTER water filters. Chiller Daddy water chillers
Large selection of replacement water filters, reverse osmosis membranes, testing kits and water filter system repair parts. From one water filter or membrane to case quantity discounts.
LOW PRICES and FAST SHIPPING SINCE 1999 !
99.5% Of All Items In Stock For Same Day Shipping!
The PH-80 is an economically priced, reliable pH meter that is ideal for testing applications such as hydroponics & gardening, pools & spas, aquariums & reef tanks, water ionizers, drinking water and more.
pH Range: 0 - 14 Temperature Range: 1-80 °C; 33-176 °F Resolution: 0.1 pH; Temperature resolution is 0.1 °C/F Accuracy: +/- 0.2 pH; Temperature accuracy is +/-2%Calibration: Digital automatic calibration (one point) Electrode: Glass sensor and reference tube electrodes Housing: Water resistant Power source: 3 x 1.5V button cell batteries (model LR44 or equivalent; included) Dimensions: 15.3 x 3.2 x 1.8 cm (6.0 x 1.3 x 0.7 inches) Weight: 53.9 g (1.9 oz)
The sponge in my cap is dry. Is this okay?Yes. The PH-80 is shipped with the sponge dry. Prior to use, pour a small amount of the electrode storage solution (from the Mylar packet) into the sponge so that it's moist. Put the cap back on and allow the sensor to soak for 15 minutes standing upright. Additional storage solution is available for purchase.
How often do I need to calibrate?All pH meters should be calibrated anywhere from before every use to at least once a month. The frequency of calibration depends on usage and application, but you should calibrate at least once per month for proper results and a longer lifespan of your meter. A free packet of pH 7.0 buffer is included with the meter. Additional pH buffers for calibration are available for purchase. Please consult the user's guide for calibration instructions.
I use my meter for hydroponics. Does it need any special care?Yes. Nutrients in a hydroponics reservoir will quickly adhere to the sensor and throw off readings. It is very important to rinse the sensor in distilled water after each use. Otherwise, you'll just be measuring the nutrients that are already on the sensor.
pH Water Tester - Handheld